# LCR Meter Tutorial

### - the key points the LCR meter and how it can be used to measure inductance, capacitance and resistance.

LCR meters are used to measure the inductance, capacitance, and resistance of components.

The LCR meter takes its name from the fact that the inductance, capacitance and resistance are denoted by the letters L, C, and R respectively.

A variety of meters are available. Simpler versions of LCR meters provide indications of the impedance only converting he values to inductance or capacitance.

More sophisticated designs are able to measure the true inductance or capacitance, and also the equivalent series resistance of capacitors and the Q factor of inductive components. This makes them valuable for assessing the overall performance or quality of the component.

## LCR meter basics

There are several techniques that can be used for LCR meters.

• Bridge method:   This method uses a bridge which is balanced. At the balance point the bridge component positions can be used to determine the value of the component under test. This method is typically used for lower frequency measurements - often measurement frequencies of up to 100 kHz or so are used.

In bridge method uses a Wheatstone bridge configuration in which the device under test, DUT, is placed in a bridge circuit.

Basic bridge based LCR meter configuration

The DUT impedance is represented by Zu in the circuit. The impedance Z2 and Z3 are known. The oscillator circuit generally operates at frequencies up to about 100 kHz and can usually be selected before the test.

Then Z1 can be changed until no current flows through D. This is the balance position for the bridge. AT this point the four impedances in the circuit obey the equation:

This basic bridge circuit is sometimes used on its own in very primitive LCR meters. Some very old instruments actually have the elements that are manually balanced. However technology has moved on and higher levels of integration coupled with operational amplifier circuitry enable accurate automated versions of the circuit to be used.
• Current-voltage measurement:   The current voltage approach is normally used for components that are to be used for higher frequency applications. It provides a highly accurate measurement technique that can be used at high frequencies and over a wide range of values.

Often known as the RF I-V measurement method, this technique for LCR measurement uses measurements of current and voltage as the name implies. However, as the frequencies involved are high, it uses an impedance matched measurement circuit. In some cases for very high frequency and high precision measurements a precision coaxial test port may be employed.

There are two types of the voltmeter and current meter arrangements: one suited to low impedance and the other for high impedance measurements.

IV LCR Measurement for low impedance circuits

IV LCR Measurement for high impedance circuits

Using the voltage and current values from the measurements, the impedance of the device under test can easily be derived. By using a phase sensitive detector to make these measurements the relative phase of the voltage and current can be used to determine the impedance of the device under test in terms of resistance, capacitance and inductance. The inductance or capacitance and the resistance may then be displayed as separate values.

Often a transformer is used in the circuit to enable these measurements to be made and to isolate the measurements from ground. However this can limit the lower frequencies or the frequency range over which measurements can be made.

## Measurement hints and tips

In order to make the best of any measurements there are a number of hints and tips that can be taken on board.

• Measure at operational frequency:   When making measurements using an LCR meter it helps to use a test frequency as close to the actual operational frequency as possible. This means that the effects of any stray effects or changes due to frequency are minimised - for example inductor cores may have different properties at different frequencies.
• Adjust test amplitude:   In the same way that it is good practice to measure at a frequency that is as close to the operational frequency as possible, the same is true for the test amplitude. This is because component values may vary with the signal applied. This is particularly true for inductors that use cores such as ferrite that may introduce losses. These may be amplitude dependent.
• Effect of lead length:   At frequencies above 1 MHz or so the lead length can start to have an effect. As a rough guide a good estimate for lead inductance is around 10 nH per cm of lead.
• Discharge capacitors before measurement:   Some capacitors may carry a residual charge under some circumstances. It is best to discharge them before any measurements.

LCR meters are a useful test instrument. They may not be as widely used as in previous years where they were often found in goods-inwards areas for sample testing incoming components. However these days they tend to be used in some laboratories for testing the performance of components likely to be used in development as well as in may workshops where they can be used as an aid for fault finding.

These instruments are used to display the capacitance, inductance and resistance, but may also be used to measure the Q of an inductor, or the tan δ of a capacitor.

By Ian Poole

### Read more popular test equipment tutorials . . . . .

Arb / AWG Digital multimeter Oscilloscope Logic analyzer
Logic probe Function generator Frequency counter RF sig gen
Signature analyzer Spectrum analyzer RF network analyzer RF power meter
Analogue multimeter TDR LCR Meter

### In this section

Miguel Fernandez | Avnet EMEA
Too good to be true - the cost of counterfeit electronics and how to avoid them
The issue of counterfeit electronic components is one that has troubles the electronics industry - using them can have some major issues, everything from being removed from a preferred suppliers list to a reduction in quality.
Online - Effective Vector Network Analyzer Measurements
How to make effective VNA measurements at RF and microwave frequencies

More training courses

Understanding 5G
Find out all about the current thinking and the technologies likely to be used for 5G. Be prepared . . read this informed and informative white paper.

More whitepapers

##### Channels
Radio-Electronics.com is operated and owned by Adrio Communications Ltd and edited by Ian Poole. All information is © Adrio Communications Ltd and may not be copied except for individual personal use. This includes copying material in whatever form into website pages. While every effort is made to ensure the accuracy of the information on Radio-Electronics.com, no liability is accepted for any consequences of using it. This site uses cookies. By using this site, these terms including the use of cookies are accepted. More explanation can be found in our Privacy Policy